Click To View Circuit Diagram
Variable Power Supply With Digital Control
A Variable Power Supply with Digital Control is one of the most important and essential requirements in an electronics lab. The available power supply must provide variable outputs in steps. This circuit gives variable and fluctuation-free DC voltage as output in the range of 1.5V to 12V. No. of ICs used: 4
Things which you can learn:
*Application of the NE555 timer IC in the stable mode of operation.
*Application of the 3-terminal positive and negative voltage regulator ICs LM317 and LM7912 respectively.
*Use of the decade counter IC CD4017.
* Basics of Voltage and current control
A variable power supply is a must for those who are looking forward to taking their project building to the next level. Armed with this system you can have a clean DC supply without spikes and it ranges from 1.5 V to 12 V. Get ready and build this kit and enjoy making projects.
The circuit of the Variable Power Supply with Digital Control is built around an adjustable 3-terminal positive voltage regulator IC LM317, CMOS decade counter IC CD4017, timer IC NE555 and 3-terminal fixed negative-voltage regulator LM7912. The AC mains supply is stepped down by transformer X1 to deliver a secondary output of 12V-0-12V AC, 1A. The output of the transformer is rectified by a full-wave rectifier comprising diodes D1 through D4.
Capacitors C1 through C4 are connected in parallel to rectifier diodes to bypass undesired spikes and provide smooth and fluctuation-free power. Capacitors C5 and C13 are used as filters to eliminate ripple. Here both negative and positive half cycles are used to obtain positive as well as negative DC output. LED1, along with current limiting resistor R1, is used for mains ‘on’ indication. Timer IC NE555 (IC1) is wired as an astable multi-vibrator. It generates clock pulses when switch S2 is pressed.
The output of IC1 is connected, via an RC network, to the clock input of counter IC CD4017 (IC2). IC CD4017 is a decade ring counter. Each of its ten outputs goes high one by one when a clock pulse is received. The outputs of IC CD4017 are connected to the bases of transistors T1 through T10, respectively. LED3 through LED11 is used here to indicate the voltage levels.
The collectors of transistors T2 through T10 are connected to presets VR1 through VR9, respectively, which are used to set the output voltage. Adjustable voltage regulator IC LM317 (IC4) develops 1.25V nominal reference voltage (VREF) between its output and the adjustable terminal. The reference voltage appears across resistor R16. When the voltage is constant, a constant current flows through one of the output-setting variable resistors (VRset, VR1 through VR9), giving an output voltage at pin 2 of IC4 as VOUT=1.25(1+VRset/R16). Presets VR1 through VR9 is adjusted to get the desired output voltage.
The collector of transistor T1 is directly connected to ADJ terminal (pin 1) of IC4, so the output voltage of IC4 will be the voltage across fixed resistor R16, which is equal to 1.25V. When switch S3 is pressed, pin3 of IC2 goes high and the output voltage becomes 1.2V. When switch S2 is pressed, the output of IC1 goes high.
As a result, the outputs of IC2 go high one by one as a ring counter. Since presets VR1 through VR9 are connected at the collectors of transistors T2 through T10, respectively, different output resistances appear between the adjustable and ground terminals of IC4, resulting in different output voltages. By using a properly calibrated digital multimeter you can easily adjust the presets to obtain 1.5V to 12V. A fixed, negative 12V DC can be obtained by using fixed, the negative-voltage regulator IC LM7912 (IC3). Thus the power supply unit can be used for circuits requiring both negative and positive DC voltages. When CD4017 is reset by pressing switch S3, the output voltage becomes 1.2V and all the voltage-indication LEDs turn off.